When I use a custom made arbitrary wave form, set the modulation to burst and the trigger source to external it is not possible to get any signal out of the device.
This was a known limitation on AFG1062's that had firmware v1.0.4 or lower. This limitation has been resolved with the installation of firmware v1.0.5 which can be downloaded from Tek.com.
Diese FAQs beziehen sich auf:
Produktreihe Arbiträrsignal-/Funktionsgenerator AFG1000
FAQ-ID 247536Alle FAQs ansehen »